Power off the 9030 PLC. Remove the resistor, capacitor and DVM connections from the terminals. Connect the resistor,
capacitor and DVM to terminals 4 and 6 in the same manner as how they were connected to terminals 3 and 5.
Repeat steps 5 through 9 for channel 2 and P2.
Note: It may be necessary to re-calibrate P1 and/or P2 to another target other than +10.0025 V to get step #8 to pass. If so, P1
and/or P2 may be re-adjusted between +9.995 V and +10.005 V, as long as all accuracy and tolerances of step #7 are met.
12. Re-seal P1 and P2 with an anti-static potentiometer sealer.
Warning: include_once(/js/tracking.php): failed to open stream: No such file or directory in /var/www/series/revision.php on line 248
Warning: include_once(): Failed opening '/js/tracking.php' for inclusion (include_path='.:/usr/share/php:/usr/share/pear') in /var/www/series/revision.php on line 248